|(0018,9107)||MR Spatial Saturation Sequence||2 ||A sequence that provides the position of spatial saturation bands deposited as part of the pulse sequence. Zero or one Item may be included in this sequence.|
|(0018,9104)||>Slab Thickness||1 ||Thickness of slab in mm.|
|(0018,9105)||>Slab Orientation||1 ||The direction cosines of a normal vector perpendicular to the saturation plane with respect to the patient. See C.220.127.116.11.1 for further explanation.|
|(0018,9106)||>Mid Slab Position||1 ||The x, y, and z coordinates of the midpoint of the slab plane in mm with respect to the patient. See C.18.104.22.168.1 for further explanation.